Can't turn off Agilent detector lamp with Empower 3 and Agilent ICF 2.1 Hot Fix 1
We are in the process of upgrading to Empower 3 SR2 HF2. When testing the Agilent ICF 2.1 HF1 ICS for shutdown instrument method of Agilent 1100 LC, we following the release notes by enabling "Shutdown after Run" option under General tab, and entered 1 minute as the Stop time of the Quat. Pump, Flow rate 0 mL/min, VWD Stoptime is "As Pump/Injector" (default value). When we run the sequence, we added the last row in sample set as condition column with the method set using the shutdown method above with 1 minute run time, but the run continues going above 1 minutes and has to be manually aborted after 10 minutes. The lamp is also still on. For those of you using the Agilent ICF 2.1 HF1 under Empower 3, is there anything else we need to do to enable the lamp off function in Empower 3?