Message center errors
I have a remote facility that I manage that is generating an unusual amount of errors in the message center. The site has about 10 users of Empower and 6-7 instruments. The message center generates about 15,000 entries per week. This is a vanilla Empower 3 install.Clearly, I can't even begin to look through all of these errors as there are simply too many of them. However, the four most prevalent are:1. G1314A: signal run buffer full. Storing of signal data stopped.2. G1314A: Extract Data: Software has failed to append raw data to the instruments ring buffer.3. Oracle Error Error - OCI_INVALID_HANDLE4. Oracle Error ORA-00904: "PEAK"."ATM6_IMP_": invalid identifier.I can find next to no reason/explanation for any of these. Clearly, some are resulting from Agilent systems. Additionally, at an unrelated site and unrelated version of Empower (Empower 2 FR4) where we are also using Agilent systems... I have one system that simply stops. No error is given in the message center, no errors are listed in the sample audit trail. When you connect to the instrument the bottom status bar shows "instrument failure", yes none of the stacks in the system have a red status and the system is still pumping/heating/cooling like noting ever want wrong.Can anyone (perhaps people from Waters) shed some light as to what is going on here? My prevailing idea of gremlins in the system gives staff a good laugh, but I really think we would all like to stop laughing about this and solve the problems. Any help is greatly appreciated.