How do I connect a Waters fraction collector III to my UPLC Acuity system for software control?
<p> How do I connect a Waters fraction collector III to my UPLC Acuity system (outfitted with Acquity autosampler and PDA) in such a way that I can program the fraction collector via Fractionlynx software? I expected to see an ethernet plug on the fraction collector that would connect to the sample manager, but the fraction collector does not have an ethernet port.</p><p>Will I be using the stripped wire ends of an unspecified cable to plug into ports 3 & 4 of the fraction collector I/O terminal? Then do I connect these same wires to the corresponding terminals on my <em><strong>detector</strong></em> or to the <em><strong>sample manager</strong></em>, terminal I or II on each, or what? I haven't found any instructions that are very clear to me.</p><p>Thanks</p>